Large Screen Blood Glucose Meter
Test time sample volume: 5 seconds up to 0.1 ul
Battery one: 3.0v CR2032 Lithium battery
Battery life: About 1000 tests
Large Screen Blood Glucose Meter Introduction
Blood glucose meter, is a kind of electronic instrument for measuring blood sugar level. Blood glucose measurement is a kind of blood test, which can be measured by whole blood, serum or plasma samples of glucose concentration. The main test methods for chemical and enzymatic.
How to Use
First, insert a test strip to turn on the meter.
Second, hold the lancing device firmly against your finger.
Third, press the release button.
Fourth, hold the tip of the test strip to the drop of blood.
Fifth, after 5 seconds, the results will appear.
Parameter of Large Screen Blood Glucose Meter:
|Range of Glucose Measurement Sample Type Calibration against||40-600mg/dl(2.2-33.3mm0l/L) Capillary whole blood plasma-equivalent|
|Test time sample volume||5 seconds up to 0.1 ul|
|Battery||one 3.0v CR2032 Lithium battery|
|Battery life||About 1000 tests|
|Unit of measure memory||mmol/L or mg/dl switchable 360 results with date and time|
|Hematocrit automatic power off||30%-50% Approx 5 minutes after last user action|
|Mini order payment||200pcs T/T,L/C,Western Union, PayPal|
If the error of the fast blood glucose meter is within 20%, it is generally considered that it will not affect the change of the treatment measures and is therefore acceptable. Therefore, the World Health Organization's requirements for blood glucose meter error are within 20%.
The National Standard for Blood Glucose Meters issued by the General Administration of Quality Supervision, Inspection and Quarantine of China, GB/T 19634-2005, stipulates that the blood glucose meter is qualified according to the following requirements:
When the test range is less than or equal to 4.2mmol/L, the allowable deviation does not exceed plus or minus 0.83mmol/L.
The test range is greater than 4.2mmol/L. The allowable deviation is no more than plus or minus 20%.
The difference between the batches of blood glucose test strips should be no more than 15% between different batches of blood glucose test strips.
Our company is one of the leading large screen blood glucose meter manufacturers and suppliers in China for our quality products. Please feel free to wholesale bulk large screen blood glucose meter here from our factory. Customers are welcomed to raise any question, pls feel free to contact us, we will reply within 24 hours.